Have questions? Contact Us
A502 sensors are square standard sensors with a 2-pin male connector. They have a force range of 0-222 N (0-50 lb), specified with Tekscan electronics. The model is linear through a much lower range of 0-22N (0-5 lb), and is capable of measuring loads up to 44,482 N (10,000 lb).*
The dynamic range of this versatile force sensor can be modified by changing the drive voltage and adjusting the resistance of the feedback resistor (see "How to Adjust the Force Range" within the Electronics tab on the left).
This is our largest standard sensor.
Linearity (Error): < ±3% of Full Scale (Line drawn from 0 to 50% load)
Repeatability: < ±2.5% (Conditioned Sensor, 80% of Full Force Applied)
Hysteresis: < 4.5 % of Full Scale (Conditioned Sensor, 80% of Full Force Applied)
Drift: <5% / logarithmic time (Constant Load of 111 N (25 lb))
Response Time: < 5 µsec (Time required for the sensor to respond to an input force; Impact load - recorded on Oscilloscope)
Operating Temperature: -40°C - 60°C (-40°F - 140°F)††
Acceptance Criteria: ±40% sensor-to-sensor variation
††Force reading change per degree of temperature change = 0.36%/ºC (±0.2%/ºF)
If your application cannot allow an Op Amp Circuit, consult the FlexiForce Integration Guide, or contact a FlexiForce Applications Engineer. If you prefer to build your own electronics, we recommend using a circuit similar to the diagram below. This will provide linear data, making it easier to calibrate the sensor.
*In order to measure higher forces, apply a lower drive voltage (-0.5 V, -0.25 V, etc.) and reduce the resistance of the feedback resistor (1kΩ min.) To measure lower forces, apply a higher drive voltage and increase the resistance of the feedback resistor.
The FlexiForce OEM Development Kit allows design engineers to economically obtain proof-of-concept data while determining how FlexiForce force sensors will behave within their product or application environment.
The FlexiForce Quickstart Board is a finished single voltage source circuit which can be quickly dropped into a prototype or easily designed into a product to obtain force measurements.
To print this web page, please use our "share" tools.