Have questions? Contact Us
The A401 has a 2-pin connector. This is our standard sensor with the largest sensing area: 25.4 mm (1 in.). A401 sensors are available in one force range: 111 N (0 - 25 lb).* The force ranges stated are approximations.
The dynamic range of this versatile force sensor can be modified by changing the drive voltage and adjusting the resistance of the feedback resistor (see "How to Adjust the Force Range" within the Electronics tab on the left).
Sensor output is a function of many variables, including interface materials. Therefore, Tekscan recommends the user calibrate each sensor for the application. The graph below is an illustration of how a sensor can be used to measure varying force ranges by changing the feedback resistor (the graph below should not be used as a calibration chart).
|Sensor Thickness:||0.203 mm (0.008 in.)|
|Force Sensor Length:||56.9 mm|
|Sensing Size (Diameter/Width):||25.4 mm|
|Temperature Range (Low):||-40 °C|
|Temperature Range (High):||60 °C|
|Connector:||2-pin Male Square Pin|
|Pin Spacing:||2.54 mm (0.1 in.)|
|Notes:||Tekscan's largest circular FlexiForce standard sensor.|
|Linearity (Error):||< ±3% of Full Scale (Line drawn from 0 to 50% load)|
|Repeatability:||< ±2.5% (Conditioned Sensor, 80% of Full Force Applied)|
|Hysteresis:||< 4.5 % of Full Scale (Conditioned Sensor, 80% of Full Force Applied)|
|Drift:||<5% / logarithmic time (Constant Load of 111 N (25 lb))|
|Response Time:||< 5 µsec (Time required for the sensor to respond to an input force; Impact load - recorded on Oscilloscope)|
|Operating Temperature:||-40°C - 60°C (-40°F - 140°F)|
±40% sensor-to-sensor variation
≥ 3 million actuations (Perpendicular load, room temperature, 22 N (5 lb))
0.36%/°C (± 0.2%/°F) (Conductive heating)
If your application cannot allow an Op Amp Circuit, consult the FlexiForce Integration Guide, or contact a FlexiForce Applications Engineer. If you prefer to build your own electronics, we recommend using a circuit similar to the diagram below. This will provide linear data, making it easier to calibrate the sensor.
*In order to measure higher forces, apply a lower drive voltage (-0.5 V, -0.25 V, etc.) and reduce the resistance of the feedback resistor (1kΩ min.) To measure lower forces, apply a higher drive voltage and increase the resistance of the feedback resistor.
The FlexiForce OEM Development Kit allows design engineers to economically obtain proof-of-concept data while determining how FlexiForce force sensors will behave within their product or application environment.
The FlexiForce Quickstart Board is a finished single voltage source circuit which can be quickly dropped into a prototype or easily designed into a product to obtain force measurements.
To print this web page, please use our "share" tools.